medhatg wrote:First, thank you for everyone's wonderful support here.
2 days ago, I got activated by my doctor 3 weeks post op. I'm implanted with LGX 700(24+4). The surgery and the healing went nice. Here comes the important part, unfortunately, I cannot pump to the full hard erection all of you guys are talking about. The pump is like a piece of rock and all I get is a limp dick. I tried the following:
- Completely deflate, the quickly press the pump to get the "pop".
- Sit in the bathtub and try to work-out the pump.
- Wait for few minutes, then try pumping again.
Any ideas, advices?
Thank you in advance
There is no such thing as LGX 24cm; the longest LGX is 21cm. Believe me, I was tried to be sold as “specially made for me” size, but it has been more than debunked that it does exist. So, if you have an AMS 24cm, it is a CX, not a LGX.